ACCU-CHEK AVIVA NANO MANUAL PDF
Unlike some systems that require you to scroll through multiple numbers to code the meter manually, the code chip for the Accu-Chek Aviva Nano blood glucose. View and Download Accu-Chek Aviva Nano user manual online. Aviva Nano Blood Glucose Meter pdf manual download. Page 1. Page 2. Page 3. Page 4. Page 5. Page 6. Page 7. Page 8. Page 9. Page Page Page Page Page Page Page Page Page
|Country:||Bosnia & Herzegovina|
|Published (Last):||2 January 2006|
|PDF File Size:||12.6 Mb|
|ePub File Size:||2.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
You are now ready to use the Accu-Chek Connect online diabetes management system to communicate information from the meter. Press and release 5.
Testing Your Blood Glucose Chapter 2: The insulin pump should be approximately 4 inches 10 cm from the infrared cable. Accu-Chek Aviva Nano Click for instructions. Accu-Chek Guide Click for instructions.
Symptoms of High or Low Blood Glucose Being aware of the symptoms of high or low blood glucose can help you understand your test results and decide what to do if they seem unusual.
Only test results that have been assigned manuak pre-meal or post-meal marker are included in pre-meal and post-meal averages.
ACCU-CHEK AVIVA NANO USER MANUAL Pdf Download.
Trigger the lancing blood drop symbol firmly against a fleshy device while keeping appear on the display. Do not put blood on top of the test strip.
The built-in infrared window at the bottom of the Accu-Chek Spirit insulin pump allows for communication between the insulin pump and the computer. Page 9 Golden End — Insert this end of the test strip into the meter. Doing so puts the Accu-Chek Compact meter in communication mode.
Devices Available Outside the United States
Perform a ensure the time and date in blood glucose or control test. You should perform a control test when: Page 4 It should not be used to aviav diabetes. Got it, continue to print. The Accu-Chek Performa Insight diabetes manager, also referred to as a meter, communicates with the Accu-Chek Connect online diabetes management system through a USB cable that connects the meter to the computer.
Check the USB connection. The insulin pump automatically returns to the STOP mode 15 minutes after the last data transfer when the Data Transfer screen in selected. Press and release to toggle through the test result markers and post-meal reminder.
Before attempting to connect the Accu-Chek Performa Combo meter for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:.
Accu-Chek Aviva Nano User Manual
Before attempting to connect the Accu-Chek Performa meter for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:. Accu-Chek Aviva Insight Click for instructions. Yellow Release Button 5. It should not be used to diagnose diabetes. Unusual Blood Glucose Results If your blood glucose result does not match how you feel, follow these steps: Do not remove the test strip.
To perform a blood glucose test from an alternate site, you need the meter, a test strip, and a lancing device with the clear AST cap. Select the control 4. The insulin pump should be about 4 inches 10 cm from the infrared cable.
Make sure the time and date are correct and adjust, if necessary. Accu-Chek Spirit Combo Click for instructions. Push it into the accu-chekk chip strips.
Devices Available Outside the United States
Press and release 9. A test strip and flashing 5. I need to order some self test diary but I can’t order online can you let me know how I can get some. Table of Contents Add to my manuals Add. For detailed information on how manhal obtain blood the display.
Other test strips will give inaccurate results. The meter will not turn on or the display is blank. Accu-Chek Compact Click for instructions.
Page 85 All other product names and trademarks are the property of their respective owners. Before attempting to connect the Accu-Chek Compact Plus meter for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:.
Accu-dhek Instant S Click for instructions. Why Perform Control Tests Chapter 4: Press and hold until you see the flashing test strip symbol to exit the set-up mode. Test results are stored from the newest to the oldest. Before attempting to connect the Accu-Chek Spirit insulin pump for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:. If you still have problems, contact Accu-Chek Customer Care at To use it correctly and dependably, mano need to understand its operation, screen displays and all individual features.
The software is now ready to communicate information from the insulin pump. Cacu-chek attempting to connect the Accu-Chek Compact manial for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:. The statutory provisions on rights in consumer goods sales in the country of purchase shall apply.
The meter needs little or no maintenance with normal use.